The MP Scope measuring station is a mobile shielded room (shielded tent) for sensitive measuring devices during Burst-/EFT- or RF testings at the work place of a developer. The measuring device (oscilloscope, spectrum analyzer) is put into the shielded tent. The connection to the disturbed environment is maintained via shielded SMA-SMA cables and a shielded transit within the base plate. The measuring device is powered via the filter of the base plate. It is quite common to measure with devices like an oscilloscope or a spectrum analyzer at electronic assemblies. If the measurement should be done under disturbances, there is a risk that the measuring device could be susceptible to interference. In order to prevent the interference of the measuring device it is put into the shielding tent of the MP Scope measuring station. The signals of the electronic devices to be measured are transmitted via shielded connections to the MP Scope measuring station and measuring device.
On request, the MP Scope measuring station is available with a taller working space (900 x 500 x 650) mm.
Scope of delivery
- 1x GP 23, Ground Plate
- 1x BZ 23-1, Shielding Material
- 1x ZG 23-1, Tent Poles
- 1x SMA-SMA rt 0.7 m ds, SMA-SMA Measurement Cable, red
- 1x SMA-SMA ge 0.7 m ds, SMA-SMA Measurement Cable, yellow
- 1x SMA-SMA gr 0.7 m ds, SMA-SMA Measurement Cable, green
- 1x SMA-SMA bl 0.7 m ds, SMA-SMA Measurement Cable, blue
- 1x MP Scope m, MP Scope Set User Manual
|Attenuated shielding||45 dB - 50 dB / 30 MHz - 1 GHz|
|Working surface (length x width)||(900 x 500) mm|
|Sizes (L x W x H)||(900 x 500 x 400) mm|
Scope of delivery details
The GP 23 ground plate is used as reference ground plane for EMC measurements and for shielding devices under test or corresponding measuring devices. The electric connections between the ground plate and the environment are led outward via filter or RF connections to the outside. Power supply c…
The shielding cover consists of a conductive fabric material. It is pulled over the tent poles, which are inserted into the GP 23 ground plate. Doing so a shielded room is created, which is suitable for measurements during development of electronic assemblies or larger devices. From the front, the …
These metal tent poles are designed to be a base frame for the shielding cover and are inserted into the GP 23 ground plane. It can be easily folded and needs little space for storage.