The MP ESD measuring field set is used to measure electric and magnetic fields of an ESD generator. In addition to the testing pulse, the ESD generators also emit electric and magnetic interference fields. Additionally to the test pulse these interference fields influence the device under test. These undefined influences falsify the ESD testing result.
With the MP ESD measuring field set at the working place, the developer can measure the additional ESD generator fields while changing the ESD generator position to the field measuring system.
The knowledge about intensity and local distribution of electric and magnetic interference fields of the ESD generator is an essential help for the interference supression. On request, the MP ESD measuring field set is available with a taller working space (900 x 500 x 650) mm.
Scope of delivery
- 1x BPM 02, dB/dt Field Meter
- 1x EPM 02, dE/dt Field Meter
- 1x EFM 04, E-Field Meter
- 1x GND 25, Ground Plane
- 1x GNDA 02, Ground Adapter 32.9 mm x 32.9 mm
- 1x ZG 23-1, Tent Poles
- 1x BZ 23-1, Shielding Material
- 1x GP 23, Ground Plate
- 1x SMA-SMB 1 m ds, SMA-SMB Double-Shielded Measurement Cable
- 1x SMA-SMA rt 0.7 m ds, SMA-SMA Measurement Cable, red
- 1x SMA-SMA ge 0.7 m ds, SMA-SMA Measurement Cable, yellow
- 1x SMA-SMA gr 0.7 m ds, SMA-SMA Measurement Cable, green
- 1x SMA-SMA bl 0.7 m ds, SMA-SMA Measurement Cable, blue
- 1x MP ESD field case, System Case
- 1x MP ESD Field m, MP ESD Field Set User Manual
|Frequency range||1 GHz|
Scope of delivery details
The BPM 02 is a dB/dt field meter and it is used to measure the temporal change of the magnetic field strength characteristics dB/dt up to 3 GHz. The lower limit frequency depends on the measurement set-up.
The EPM 02 is suitable for the measurement of fast transients with a bandwidth up to 3 GHz. The dE/dt field meter doesn't have a lower limit frequency. Depending on the noise limit of the used measuring device the usage of the EPM 02 is limited in the lower frequency range. For example, at a noise …
The EFM 04 field meter is used to measure fast transient fields or RF fields. Time responses or frequency responses can be detected for a bandwidth up to 3 GHz. In connection with the ICE1 IC test environment the EFM 04 is used as a measurement- and calibration bench for field sources of the IC test…
The GND 25 ground plane ensures the RF-compatible contacting of the probe during an EMC measurement at a test IC. The GND 25 consists of steel with a gold plated surface. The recess (103 mm x 103 mm) is designed for the placement of GNDA ground adapters (GNDA 01-04) and TEM cell prints (100 mm x 100 mm).
The GNDA ground adapter is designed for the adaption of the test board to the (100 x 100) mm recess of the GND 25 ground plane. The test board is placed into the recess of the ground adapter. According to the size of the test board there are four ground adapters to choose from (GNDA 01-04).
These metal tent poles are designed to be a base frame for the shielding cover and are inserted into the GP 23 ground plane. It can be easily folded and needs little space for storage.
The shielding cover consists of a conductive fabric material. It is pulled over the tent poles, which are inserted into the GP 23 ground plate. Doing so a shielded room is created, which is suitable for measurements during development of electronic assemblies or larger devices. From the front, the …
The GP 23 ground plate is used as reference ground plane for EMC measurements and for shielding devices under test or corresponding measuring devices. The electric connections between the ground plate and the environment are led outward via filter or RF connections to the outside. Power supply c…